Inicio > Noticias
Centrifuge Supplier shares with you about the maintenance of the centrifuge, hope it will be helpful to you
Maintenance of the main machine
Clean the centrifugal chamber frequently with a soft cloth or thin paper to keep the chamber clean and dry. The surface of the centrifuge and the Centrifuge chamber can be cleaned with a soft cloth moistened with neutral detergent.
To ensure proper operation of the instrument, the chamber should be kept clean and dry by wiping off the condensate after centrifugation.
The seal on the centrifuge chamber door can be cleaned every 3 months with a thin paper or soft cloth; if the chamber door O-ring is found to be worn, it should be replaced immediately.
Inspection and replacement of the tachometer disk
Before using the rotor head, you must check whether the speed dial is clean and broken, if it is not clean, you need to clean it with alcohol and wait for it to dry before using; if it is broken, you need to replace the speed dial in time.
First remove the old speed dial with tweezers or blade, clean the adhesion with acetone, then insert the tool, place the new speed dial along the tool to the bottom of the rotor head and press. Remove the tool, place the speed dial face down on the head and press it for at least 2h before using the head
We also offer Screen Centrifuges and decanters, please feel free to contact us if you need them.